The very high temperatures found inside steam boilers (in excess of 1500 degrees Fahrenheit, or 800 degrees Celsius) can cause the formation of a specific type of hard, brittle iron oxide called magnetite on the inside and outside surfaces of steel boiler tubing. The presence of this oxide layer on the outside of tubes can interfere with ultrasonic wall thickness measurements using common dual element transducers, both because it can have a very rough surface that prevents proper sound coupling, and because the oxide thickness is added to the steel thickness. However, as its name implies, magnetite is magnetic, and that property permits the use of magnetostrictive EMAT (electromagnetic acoustic transducer), such as the Panametrics-NDT E110-SB. EMATs offer several advantages over conventional dual element piezoelectric transducers: the need to remove oxide scale is eliminated, the scale thickness is not added into the wall thickness measurement, and measurements can be made very quickly and without the need for liquid couplant. The primary limitation of magnetostrictive EMATs is that they only work when scale is present and bonded to the outside of the boiler tube. Additionally, the minimum measurable wall thickness and measurement accuracy is not nearly as good as what can be achieved with a conventional dual, and EMATs are relatively insensitive to small internal pits. For these reasons, EMATs are often used for quick initial wall thickness surveys, while duals can then be used to take a closer look at areas of concern.
Theory of operation
Figure 1 - Cross section of a typical EMAT
Figure 2 - Method of sound wave generation
Because the scale itself is the transducer element, scale roughness is not a coupling issue, and the scale is not added to the thickness measurement. The EMAT generates a shear wave, so the instrument must be calibrated to a shear wave velocity of approximately 0.1280 in./µs (3,240 m/s) in typical carbon steel. Typical measurement accuracy with the E110-SB EMAT transducer is ± 0.010 in. (0.25 mm), with a minimum measurable thickness of at least 0.080 in. (2.0 mm) depending on material properties.
Setup and measurement procedures
The quality of ultrasonic echoes in EMAT applications depends, in part, on the consistency of the oxide scale layer, which can vary from point to point on a given boiler tube. If usable echoes cannot be obtained at one point, try another point nearby. Also, the E110-SB transducer incorporates an adjustable standoff that varies the distance between the transducer face and the surface of the boiler tube. Adjusting this standoff distance helps optimize the echo response in many cases.
(a) 38DL PLUS® gauge: The E110-SB transducer is used with the 38DL PLUS thickness gauge in conjunction with a 1/2XA/E110 adaptor, which provides both probe recognition and the added high-pass filtering that is necessary for proper signal conditioning. When the adaptor is plugged in, the gauge will automatically selects the EMAT default setup DEFM1-EMAT/E110. As with any gauge setup, for optimum accuracy, a two-point velocity/zero calibration should be performed on thick and thin reference standards of known thicknesses. If such standards are not available, the default settings are usually sufficient as a starting point. Instrument gain and echo blanking may be adjusted as necessary to optimize echo detection. Under the EMAT default setup, the 38DL PLUS gauge will display a full wave rectified waveform. A typical waveform display is seen below in Figure 3.
Figure 3 - Typical rectified EMAT waveform
Figure 4 - Typical RF EMAT waveform
(b) EPOCH 6LT or EPOCH 650: A typical EMAT starting setup and waveform for an EPOCH series flaw detector is seen in Figure 5. Note that bandpass filtering must always be used to filter out low frequency noise associated with EMATs. Also, because shear wave frequency varies with oxide scale thickness, the EPOCH's square wave frequency should be adjusted as necessary to optimize the echo response.
Figure 5 - Basic setup parameters: velocity, 0.1280 in./µs; zero, 0.650 µs; energy, 400 V; damping, 50 Ω; pulse frequency, 5.0 MHz; filter, 1.5-8.5 MHz
Products used for this application
EMATs (electromagnetic acoustic transducers) are single element transducers that use a magnetostrictive effect to transmit and receive ultrasonic waves. EMATs offer unique advantages—they can be used without couplant, there’s no need to remove external scale prior to an inspection, they can be used in contact or at a small distance from the surface, they can be used on hot parts, and they work well on small-diameter pipes.
The EPOCH 650 is a conventional ultrasonic flaw detector with excellent inspection performance and usability for a wide variety of applications. This intuitive, rugged instrument is a continuation of the popular EPOCH 600 flaw detector with additional capabilities.
The 38DL PLUS is an advanced ultrasonic thickness gauge. Uses dual element transducer for internal corrosion applications, and has features that include THRU-COAT technology and echo-to-echo. Uses single element transducers for very precise thickness measurements of thin, very thick, or multilayer materials.
Source : https://www.olympus-ims.com/en/applications/thickness-boiler-tubes-emat-transducers/